Hearing Loops in the news
Induction Loop Systems
Induction Loop Receivers
Induction Loop Testing Equipment
Multisensory Sound Lab
Loop Systems & Accessories
Pricing & Ordering
Calibrating Induction Loop Systems to IEC 60118-4:
Frequency response: 100-5000Hz ±3dB relative
Field strength: 400mA/m peak response
“Listening area” has been replaced with “listening volume”, i.e. listening
height has to be integrated into the specification. This means that the
magnetic field should be homogenous (±3dB) within the listening area and
at a stated minimum and maximum height approximating seated and standing
positions, for example 1.2m (seated) and 1.7m (standing).
Specified reference level: 400mA/m=0dB. A certifiable loop system must
achieve 400mA/m in program peaks.
Background noise is measured with the Field Strength Meter (FSM) in
A-filter position. The following is valid for the signal/noise (S/N) ratio
relative to 400mA/m=0dB:
- 47dB S/N ratio is desirable with loop amplifier connected
- If S/N ratio is lower than 32dB it should be reported and measures to
overcome the problem discussed with the loop installer/manufacturer
- For transitory listening (for example across-the-counter systems) a S/N
ratio down to 22dB can be accepted
Before an induction loop system is permanently installed, we recommend:
- Environmental electromagnetic noise should be checked with the venue’s
lights and all electrical systems turned on.
- The loop wire should be temporarily taped or tacked in place and the
system fully tested before performing final installation.
- To check whether the loop system is interfering with audiovisual
systems, the venue’s audio & video systems should be on when testing the
- If electric guitars are used, they should be checked during the
pre-installation loop test in order to determine if their magnetic pickups
are sensitive to the planned loop location. For installation guidelines,
please click here.
- We have many satisfied customers who simply use our Induction Loop
Receiver (included with the Satellite III) to set up their systems. The
receiver provides a gross visual LED indication of proper field strength
and the installer may check sound quality and noise using the supplied
- In order for a looped site to be properly certified as meeting
international standards, a calibrated field strength meter kit, such as
our FSM Loop Test Set must be used.
Back to Induction Loop Assistive
Systems' List, click here.
Oval Window Audio Today
If you have any questions about calibrating induction
simply e-mail us directly. One of our representatives will get back to
you promptly. Just e-mail your message to email@example.com
Home Page // Hearing Loops // Induction Loop Assistive Listening
// Induction Loop Receivers // Induction
Loop Testing Equipment // Vibrotactile/Multisensory
Sound Lab // Loop Systems & Accessories Pricing &
Ordering // Contact Us
This site created with the help of The Write Direction